It is packaged in a case, which allows its use in workshop and in situ (aboard ships for example). It operates with free air, a phosphine cylinder (PH3) and a sulfur dioxide cylinder (SO2), used for organophosphorus (G, Vx) and organosulfur (HD) simulation.
It is fitted with a diffuser joined with the bench by a flexible hose. The diffuser is used on the case for the AP2C checking, or remote to check AP2C-V or ADLIF if located on a vehicle or on a ship.
The bench is supplied with:
The air required to dilute SO2 and PH3 gas is taken directly from the surrounding air and is fed to the diffuser by a vane pump. The air filter eliminates all traces of dusts and water vapor.
A mass flow meter measures the actual airflow to allow the SO2 and PH3 flows to be set.
Two mass flow meters check the SO2 and PH3 flows, before reaching the mixer, which mixes the gas and air required to simulate toxic agent for the devices calibration.
Concentrations of generated gases:
The bench can be joined to a Portable Terminal (TP) or to a PC type microcomputer in order to visualize the different detectors internal parameters and to calibrate them.
Weight: 20 kg
External Size (LxWxH): 550 mm x 420 mm x 320 mm (21.65 ” x 16.53 ” x 12.59 ”)
Supply voltage: 100 to 240 Vac 50/60 Hz
Operating temperature: +10°C to +40°C (+50°F to +104°F)
Storage temperature: +0°C to +50°C (+32°F to +122°F)